|Screen||19-inch monitor with a minimum resolution of 1366x768 with 16 bit color depth.|
|Mouse||Mouse with a central scroll wheel.|
|USB||Minimum 1 USB port to insert the protection key.|
|Compatibility||“BTI Scan® 3” is charaterised by being an open platfom compatible with different CT scanning systems; conventional, spiral, volumetric, etc. that provide a study of the patient's maxillary in DICOM format. It also has an extensive library of implants available to the user for pre-surgery planning on the patient's scan. In addition, it allows you to import studies created with previous versions of “BTI Scan®”.|
Configuration of CT Scan Paramaters
Parameters for Helical CT scan with “BTI Scan® 3” Sequence of Axial Sections
The sections must be equal and homogenous (if they are not, the “BTI Scan® 3”, program, when carrying out the diagnosis and simulation, will display them as erroneous and will mark them in black; if the ratio of valid sections/Non-valid sections is higher than 20%, “BTI Scan® 3” will not load the CT scan and will declare it as non-valid.
The thickness of sections must be 1mm maximum; the lower the distance between sections, the higher the quality when viewing them. “BTI Scan® 3” supports submillimetric distances of up to 0.6mm.
|Kilovoltaje (Kv)||120 – 160 Kv|
|Milliamperage (mA)||90 – 140 mA|
|FOV (Field of View)||12 – 18 cm|
|SFOV (if available)||HEAD|
|Matrix||512 = (512*512)|
IMPORTANT WARNING REGARDING CONE BEAM OR VOLUMETRIC SCANNERS
The reliability of the data and measurements provided by DICOM images obtained with CONE BEAM or VOLUMETRIC scanners can vary depending on the technique, Energy parameters and the equipment used.
|Space between sections|
|Superior maxilla||0,5 – 1,0 mm|
|Inferior maxilla||0,5 – 1,0 mm|
|Helical thickness||1,0 – 1,5 mm|
|Exploration mode||(0,0 – 1):1|
|Speed (mm/rot)||(0,8 – 1,5)|
|Bone or High Resolution.|